- Open Access
Enhancement of mechanical strength and in vivo cytocompatibility of porous β-tricalcium phosphate ceramics by gelatin coating
© Furusawa et al. 2016
- Received: 22 October 2015
- Accepted: 2 February 2016
- Published: 6 February 2016
In an attempt to prepare scaffolds with porosity and compressive strength as high as possible, we prepared porous β-tricalcium phosphate (TCP) scaffolds and coated them with regenerative medicine-grade gelatin. The effects of the gelatin coating on the compressive strength and in vivo osteoblast compatibility were investigated.
Porous β-TCP scaffolds were prepared and coated with up to 3 mass% gelatin, and then subjected to thermal cross-linking. The gelatin-coated and uncoated scaffolds were then subjected to compressive strength tests and implantation tests into bone defects of Wistar rats.
The compressive strength increased by one order of magnitude from 0.45 MPa for uncoated to 5.1 MPa for gelatin-coated scaffolds. The osteoblast density in the internal space of the scaffold increased by 40 % through gelatin coating.
Coating porous bone graft materials with gelatin is a promising measure to enhance both mechanical strength and biomedical efficacy at the same time.
- Compressive strength
Porous hydroxyapatite (HA) and β-tricalcium phosphate (β-TCP) have long been clinically used for bone grafts because they enable perfusion of cells and other factors necessary for bone regeneration and because they allow bone ingrowth [1–3]. There have been a large number of works on the effect of porous structures on biological efficacy. For example, pores larger than 100 mm are essential for bone ingrowth into HA scaffolds [4, 5], and larger pores facilitate faster bone ingrowth . Therefore, much effort has been devoted to the fabrication of scaffolds with pores as large as possible, and various processing techniques have been reported, although most of them were attempted on HA [7, 8]. Examples include freeze casting [8–10], sponge templating [11, 12], gel casting , particle templating [14, 15], whisker sintering , robocasting , extrusion deposition , and slurry foaming .
Although larger pores and porosities are favorable for faster bone ingrowth, they deteriorate the mechanical strength of the scaffolds. From a practical viewpoint, the compressive strength needs to be higher than approximately 1.0 MPa in order to avoid collapsing of the scaffolds during the handling for implantation. Therefore, there is a trade-off between the mechanical strength and the porosity. One of the promising methods to reinforce scaffolds without lowering its porosity or biomedical efficacy is to coat the scaffolds with biocompatible polymers because the infiltration of polymers into the microcracks of the scaffolds reduces the fracture origin . The most widely used polymers are poly(lactic acid)- and poly(caprolactone)-based polymers [14–16, 21–25]. Others include glycerol sebacate , gelatin [14, 27], and collagen . When scaffolds were coated with poly(lactic acid) or poly(caprolactone), the compressive strength generally increased [21–25]. However, the effects of those coatings on the biomedical efficacy of the scaffolds differ from report to report. In some cases, in vitro differentiation of MC3T3-E1 preosteoblast cells and bone marrow stroma cells were promoted , whereas in other cases, the initial attachment and proliferation were suppressed [16, 23, 26]. Biomedical evaluation was not quantitatively described in some reports [22, 25]. Some studies reported the in vitro cytocompatibility of collagen-coated scaffolds, in which the proliferation and differentiation of MG63 were enhanced  and the differentiation of rat-originated osteoblasts was promoted . Still, in vivo evaluation of those polymer-coated scaffolds has been very scarce.
One of the present authors (SY) has developed endotoxin-free gelatin for regenerative medicine . Gelatin is an inherently cytocompatible substance, and its mechanical strength can be enhanced by thermal cross-linking. Therefore, gelatin is a promising substance for the reinforcing coating of porous ceramic scaffolds.
In the present work, we first reinforced porous β-TCP scaffolds by gelatin coating, followed by thermal cross-linking. Then, the resultant scaffolds were evaluated for in vivo cytocompatibility from animal implantation tests.
Preparation of porous β-TCP blocks
Porous β-TCP scaffolds were prepared in our laboratory by sintering porous green bodies in the following manner. Commercial β-TCP powder (β-TCP −100, Taihei Chemical Industrial Co., Ltd., Osaka, Japan) was ground with an automatic agate mortar for 30 min to crush any coarse agglomeration. Then, 36.84 g of the β-TCP powder was added to the dispersion medium, which was prepared by dissolving 0.325 g of polyvinyl alcohol (polymerization degree 2000) and 3.0 g of an ammonium polyacrylate-based dispersant (Kaocera 2000, Kao Corp., Tokyo, Japan). The mixture was ball milled for 12 h. MgO (0.37 g) was added to suppress the phase transition during sintering, and the mixture was ball milled again for 1 h to prepare a well-dispersed slurry. The solid content of the slurry was approximately 45 vol%.
A foaming agent (6 mL, EMAL D-3-D, sodium polyoxyethylene alkyl ether sulfate, Kao Corp.) was added to 30 g of the slurry. The mixed slurry was then whisked with a kitchen blender. The whisked slurry was poured into a polymer mold approximately 40 × 40 × 50 mm in volume, frozen with liquid nitrogen vapor, and then lyophilized to give a porous green body. The green body was sintered at 1473 K for 12 h in ambient air to obtain β-TCP scaffolds. The porosity of the as-sintered scaffolds was 92 % as measured by the Archimedes method.
Two kinds of gelatin were used: reagent-grade gelatin (Wako Pure Chemicals Ind., Ltd.) for the preliminary experiments and regenerative medicine-grade gelatin (RM-100, Jellice Co., Ltd.) for the final experiments. Porous β-TCP scaffolds were immersed in aqueous solutions containing 0.5, 1.0, or 2.0 mass% gelatin for 30 s, taken out, and the redundant solution was removed by wiping the blocks with paper towels. The β-TCP blocks bearing gelatin solutions were cooled in a refrigerator at 253 K overnight and then dried at room temperature in a vacuum. The dried, gelatin-coated β-TCP scaffolds were subjected to heat treatments in a vacuum to obtain cross-linked gelatin. The cross-linking temperatures were 373, 393, 413, and 433 K, and the duration was 12 h.
Characterization of gelatin-coated β-TCP
The crystalline phase of the sintered scaffold was ascertained to be β-TCP by X-ray diffractometry (data not shown). The gelatin content was determined by thermogravimetry. The compressive strength of the β-TCP scaffolds, both gelatin-coated and uncoated, using samples approximately 10 × 10 × 20 mm in size was measured on an Aikoh testing machine at a crosshead rate of 1.0 mm/min. The compressive strength was defined as the maximum stress before the strain exceeded 10 % of the specimen length. The microstructure was observed with a scanning electron microscope (SEM, e-SEM, Shimadzu Rika Corp., Tokyo, Japan).
Animal implantation test
Preparation conditions and physical properties of gelatin-β-TCP scaffolds
Gelatin content (mass%)
Cross-link temp (K)
Average compressive strength σ (MPa)
Physical properties of the β-TCP scaffolds
Table 1 summarizes the preparation conditions and physical properties of the gelatin–β-TCP scaffolds. The contents of the gelatin coatings varied from 0.6 to 3.0 mass%, depending on the gelatin concentration in the coating solution. The decrease in porosity after the gelatin coating was small, and all the scaffolds had porosities higher than 90 %.
In vivo tests
The fracture starts when σ m exceeds the theoretical strength of the material, σ th. Therefore, the mechanical strength of a material increases as the cracks become less sharp and shallow. As shown in Fig. 3, the coated gelatin seemed to infiltrate into the microcracks of the framework of the scaffolds and flatten the framework surface, which should enhance the compressive strength of the scaffolds.
Presently, porous bone augmentation materials clinically used in Japan are fabricated so that the material possesses porosity as high as possible while retaining minimal compressive strength. To the authors’ knowledge, the lowest compressive strength of clinically used β-TCP scaffolds is 0.9 MPa (Osferion, Olympus Terumo Biomaterials, Tokyo, Japan) and its porosity is 75 %. In contrast, we have succeeded in preparing β-TCP scaffolds whose porosity and compressive strength are far higher than those of the commercial scaffolds. If the requisite minimum strength for bone augmentation material is approximately 1.0 MPa, there is a room to further increase the porosity.
In addition to reinforcing β-TCP scaffolds, the gelatin coating increased the osteoblast density near the scaffolds. This seems natural because gelatin has long been known to be a cytocompatible material. In the present study, however, 2 weeks of implantation may have been too short to observe the rates of the new bone formation within the pores of the scaffolds and of the resorption of those. Those studies will have to be conducted to more precisely and quantitatively assess the effect of gelatin coating. At least, still, a higher osteoblast density may imply faster new bone formation.
On the other hand, it has been well established that the dissolution of β-TCP promotes the migration of osteoclasts and osteoblasts , and calcium ions released from β-TCP may promote differentiation of osteoblasts [32, 33]. The gelatin coating may slow the dissolution of β-TCP, depending on the amount and thickness of the coating. Therefore, an increase in the osteoblast density alone does not guarantee fast bone regeneration. Although further studies are necessary to elucidate the biomedical efficacy of gelatin coating, this work is the first to report the in vivo effect of the gelatin coating on osteoblast density.
Hydrolysis of gelatin gives peptide oligomers. Among those, tripeptides, which consist of glycine and two other amino acids, have been proven to promote osteoblast differentiation  and in vivo bone healing [34, 35]. There is a possibility, therefore, that the coated gelatin is hydrolyzed to tripeptides, thus further promoting bone formation.
Porous β-TCP scaffolds with approximately 90 % porosity were prepared and coated with gelatin. The gelatin coating and subsequent thermal cross-linking increased the compressive strength by one order of magnitude. The highest compressive strength attained was 5.1 MPa. The gelatin-coated and uncoated scaffolds were implanted into bone defects of the cranial bones of Wistar rats for 2 weeks. The osteoblast density in the internal space of the scaffold was enhanced by 40 % by gelatin coating, implying the possibility of faster bone formation.
A part of this work was financially supported by Takahashi Industrial and Economic Research Foundation.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Hench LL. Bioceramics: from concept to clinic. J Am Ceram Soc. 1991;74:487–510.View ArticleGoogle Scholar
- Lew KS, Othman R, Ishikawa K, Yeoh FY. Macroporous bioceramics: a remarkable material for bone regeneration. J Biomater Appl. 2011;27:345–58.View ArticlePubMedGoogle Scholar
- Best SM, Porter AE, Thian ES, Huang J. Bioceramics: past, present and for the future. J Eur Ceram Soc. 2008;28:1319–27.View ArticleGoogle Scholar
- Chang BS, Lee CK, Hong KS, Youn HJ, Ryu HS, Chung SS, et al. Osteoconduction at porous hydroxyapatite with various pore configurations. Biogeosciences. 2000;21:1291–8.Google Scholar
- Ebaretonbofa E, Evans JRG. High porosity hydroxyapatite foam scaffolds for bone substitute. J Porous Mater. 2002;9:257–63.View ArticleGoogle Scholar
- Kühne JH, Bartl R, Frisch B, Hammer C, Jansson V, Zimmer M. Bone formation in coralline hydroxyapatite. Effects of pore size studied in rabbits. Acta Orthop Scand. 1994;65:246–52.View ArticlePubMedGoogle Scholar
- Sun F, Zhou H, Lee J. Various preparation methods of highly porous hydroxyapatite/polymer nanoscale biocomposites for bone regeneration. Acta Biomater. 2011;7:3813–28.View ArticlePubMedGoogle Scholar
- Fu Q, Rahaman NR, Dogan F, Bal BS. Freeze casting of porous hydroxyapatite scaffolds. II. Sintering, microstructure, and mechanical behavior. J Biomed Mater Res Part B: Appl Biomater. 2008;86B:514–22.View ArticleGoogle Scholar
- Fu Q, Rahaman NR, Dogan F, Bal BS. Freeze casting of porous hydroxyapatite scaffolds. I. Processing and general microstructure. J Biomed Mater Res Part B: Appl Biomater. 2008;86B:125–35.View ArticleGoogle Scholar
- Deville S, Saiz E, Tomsia AP. Freeze casting of hydroxyapatite scaffolds for bone tissue engineering. Biogeosciences. 2006;27:5480–9.Google Scholar
- Sopyan I, Kaur J. Preparation and characterization of porous hydroxyapatite through polymeric sponge method. Ceram Int. 2009;35:3161–8.View ArticleGoogle Scholar
- Munar ML, Udoh K, Ishikawa K, Matsuya S, Nakagawa M. Effects of sintering temperature over 1300 °C on the physical and compositional properties of porous hydroxyapatite foam. Dent Mater J. 2006;25:51–8.View ArticlePubMedGoogle Scholar
- Ramay HR, Zhang M. Preparation of porous hydroxyapatite scaffolds by combination of the gel-casting and polymer sponge methods. Biogeosciences. 2003;24:3293–302.Google Scholar
- Bakhtiari L, Rezaie HR, Hosseinalipour SM, Shokrgozar MA. Investigation of biphasic calcium phosphate/gelatin nanocomposite scaffolds as a bone tissue engineering. Ceram Int. 2010;36:2421–6.View ArticleGoogle Scholar
- Akkouch A, Zhang Z, Rouabhia M. A novel collagen/hydroxyapatite/poly(lactide-co-ε-caprolactone) biodegradable and bioactive 3D scaffold for bone regeneration. J Biomed Mater Res Part A. 2011;96A:693–704.View ArticleGoogle Scholar
- Shigemitsu Y, Sugiyama N, Oribe K, Rikukawa M, Aizawa M. Fabrication of biodegradable β-tricalcium phosphate/poly(L-lactic acid) hybrids and their in vitro biocompatibility. J Ceram Soc Japan. 2010;118:1181–7.View ArticleGoogle Scholar
- Miranda P, Saiz E, Gryn K, Tomsia AP. Sintering and robocasting of β-tricalcium phosphate scaffolds for orthopaedic applications. Acta Biomater. 2006;2:457–66.View ArticlePubMedGoogle Scholar
- Wu Q, Zhang X, Wu B, Huang W. Fabrication and characterization of porous HA/β-TCP scaffolds strengthened with micro-ribs structure. Mater Lett. 2013;92:274–7.View ArticleGoogle Scholar
- Aoki S, Yamaguchi S, Nakahira A, Suganuma K. Preparation of porous calcium phosphates using a ceramic foaming technique combined with a hydrothermal treatment and the cell response with incorporation of osteoblast-like cells. J Ceram Soc Japan. 2004;112:193–9.View ArticleGoogle Scholar
- Johnson AJW, Herschler BA. A review of the mechanical behavior of CaP and CaP/polymer composites for applications in bone replacement and repair. Acta Biomater. 2011;7:16–30.View ArticleGoogle Scholar
- Lee J, Kim IK, Kim TG, Kim YH, Park JC, Kim YJ, et al. Biocompatibility and strengthening of porous hydroxyapatite scaffolds using poly(L-lactic acid) coating. J Porous Mater. 2013;20:719–25.View ArticleGoogle Scholar
- Miao X, Tan DT, Li J, Xiao Y, Crawford R. Mechanical and biological properties of hydroxyapatite/tricalcium phosphate scaffolds coated with poly(lactic-co-glycolic acid). Acta Biomater. 2008;4:638–45.View ArticlePubMedGoogle Scholar
- Bang LT, Tsuru K, Munar M, Ishikawa K, Othman R. Mechanical behavior and cell response of PCL coated α-TCP foam for cancellous-type bone replacement. Ceram Int. 2013;39:5631–7.View ArticleGoogle Scholar
- Martínez-Vázquez FJ, Miranda P, Guiberteau F, Pajares A. Reinforcing bioceramic scaffolds with in situ synthesized ε-polycaprolactone coatings. J Biomed Mater Res Part A. 2013;101A:3551–9.View ArticleGoogle Scholar
- Zhao J, Lu X, Duan K, Guo LY, Zhou SB, Weng J. Improving mechanical and biological properties of macroporous HA scaffolds through composite coatings. Coll Surf B: Biointerf. 2009;74:159–66.View ArticleGoogle Scholar
- Yang K, Zhang J, Ma X, Kan C, Ma H, Li Y, et al. β-tricalcium phosphate/poly(glycerol sebacate) scaffolds with robust mechanical property for bone tissue engineering. Mater Sci Eng C. 2015;C56:37–47.View ArticleGoogle Scholar
- Landi E, Valentini F, Tampieri A. Porous hydroxyapatite/gelatine scaffolds with ice-designed channel-like porosity for biomedical applications. Acta Biomater. 2008;4:1620–6.View ArticlePubMedGoogle Scholar
- Lee MH, You C, Kim KH. Combined effect of a microporous layer and type I collagen coating on a biphasic calcium phosphate scaffold for bone tissue engineering. Mater. 2015;8:1150–61.View ArticleGoogle Scholar
- Optira EI, Moldovan L, Craciunescu O, Zarnescu O. In vitro behavior of osteoblast cells seeded into a COL/β-TCP composite scaffold. Cent Eur J Biol. 2008;3:31–7.Google Scholar
- Kanayama Y, Aoki C, Sakai Y. Development of low endotoxin gelatin for regenerative medicine. Biol Pharm Bull. 2007;30:237–41.View ArticlePubMedGoogle Scholar
- Kondo N, Ogose A, Tokunaga K, Ito T, Arai K, Kudo N, et al. Bone formation and resorption of highly purified β-tricalcium phosphate in the rat femoral condyle. Biogeosciences. 2005;26:5600–8.Google Scholar
- Jung GY, Park YJ, Han JS. Effects of HA released calcium ion on osteoblast differentiation. J Mater Sci Mater Med. 2010;21:1649–54.View ArticlePubMedGoogle Scholar
- Maeno S, Niki Y, Matsumoto H, Morioka H, Yatabe T, Funayama A, et al. The effect of calcium ion on osteoblast viability, proliferation and differentiation in monolayer and 3D culture. Biogeosciences. 2005;26:4847–55.Google Scholar
- Tsuruoka N, Yamato R, Sakai Y, Yoshitake Y, Yonekura H. Promotion by collagen tripeptide of type I collagen gene expression in human osteoblastic cells and fracture healing of rat femur. Biosci Biotechnol Biochem. 2007;71:2680–7.View ArticlePubMedGoogle Scholar
- Hata S, Hayakawa T, Okada H, Hayashi K, Akimoto Y, Yamamoto H. Effect of oral administration of high advanced-collagen tripeptide (HACP) on bone healing process in rat. J Hard Tissue Biol. 2008;17:17–22.View ArticleGoogle Scholar