- Short Report
- Open Access
Performance of the counter-torque technique in the explantation of nonmobile dental implants
International Journal of Implant Dentistry volume 6, Article number: 1 (2020)
The application of the counter-torque technique has been proposed as a conservative and atraumatic alternative for the explantation of nonmobile dental implants. The objective of this report is to assess the performance of this technique in a large number of patients.
Three hundred and fifty-five patients were treated for the explantation of 749 nonmobile dental implants. The explantations were performed by the application of counter-torque to break the bone-implant interface. Successful implant explantation was achieved in 98.4% of the implants. The frequency of complications was 1.3%, most commonly related to the appearance of fissure lines at the implant neck.
The counter-torque technique has a high success rate but is not exempt from complications, although at a very low rate.
Although presenting high and acceptable survival rates [1, 2], dental implants fail as a consequence of infection (periimplantitis), excessive biomechanical stress, or improper positioning. Different techniques have been described to remove a failed dental implant that include block resection, buccal bone osteotomy, trephine osteotomy, and piezosurgery [3, 4]. The application of counter-torque to break the implant-bone interface has been proposed as a safe, efficient, and atraumatic strategy to remove nonmobile implants . As a minimally invasive technique, Solderer et al. have recommended the counter-torque technique as the first option for the removal of failed nonmobile dental implants . Nevertheless, there is a need of studies with a larger sample size to confirm the efficacy and safety profile of this strategy. This study aims to assess the performance of the counter-torque strategy for the removal of a large number of nonmobile dental implants.
Patients of legal age treated for implant explantation of nonmobile dental implants between March 2010 and December 2018 were included in this retrospective study.
The treatment was performed using an implant removal kit (BTI Biotechnology Institute, Vitoria, Spain) that allowed the application of a counter-torque to the bone-implant interface . The decision to raise a flap or not was made according to the surgical needs. Following the manufacturer instructions, a ratchet was engaged into the implant connection and then a counter-torque was exerted by a wrench in counter-clockwise direction. If the counter-torque exceeded the 200 Ncm limit (the torque wrench opened), trephine bur was used to cut into the first 3–4 mm of implant-bone contact. The implant explantation was then continued with the torque wrench. Treatment was considered a failure if the implant could not be removed by this procedure. All the complications occurring during the procedure were also recorded.
The cause of implant removal was grouped into biological (excessive bone loss), mechanical (improper implant positioning, prosthetic failure), and surgical (treatment of lesion that involve dental implant like medication-related osteonecrosis of the jaw).
The bone level around the implant before the explantation was measured on panoramic radiographs (Sidexis XG, Sirona Dental Systems). The percentage of the implant length covered by bone was measured mesially and distally to the implant by considering the implant length as the 100%. Then the mean of the two measurements was calculated. The frequency of the implant covered by bone was grouped into four different categories 0–25%, 26–50%, 51–75%, and 76–100%. It was not possible to calculate the implant length covered by bone in millimeters as the implant length was unknown for most of the implants.
The frequency of the variables of interest (implant location, cause of implant removal, implant length covered by bone, implant removal success, and complications) was calculated and represented in percentage. Statistical analysis of categorical variables was calculated using the Chi-square test. The IBM SPSS Statistic v15 software package (SPSS Inc., Chicago, IL, USA) was used. Statistical significance was set at p value < 0.05.
In this report, 749 nonmobile dental implants were explanted in 355 patients. Figure 1 shows the anatomical location of the dental implants. The 50.6% of the removed implants were located in the maxilla and 48.2% of the cases were located in the anterior areas. Attending to the causes of explantation, the vast majority of the explantations occurred due to biological complications (86.2%), followed by mechanical complications (11.9%) and surgical intervention (1.9%). Figure 2 shows the distribution of the dental implants according to the cause of implant removal.
The bone level of the explanted implants was homogenously distributed among bone level categories. In this sense, 26.4% of the implants were included in the 0–25% of bone level category, 24.3% in the 26–50% of bone level category, 26.9% in the 51–75% of bone level category and, finally, 22.4% of the explanted implants were in the 76–100% bone level category. The implants were completely covered by bone (100%) in the 76–100% of bone level category were removed due to mechanical problems.
Regarding the success rate, 98.4% of the dental implants were successfully removed following the described procedure. Among these successful explantations, the use of specialized trephine burs was needed in 7.5% of the cases. It was studied if the bone level was associated with the use of trephine burs. There was no statistically significant (p = 0.249) relationship between trephine bar use and bone level.
The extraction kit also demonstrated to cause minimal complications that occurred in 1.3% of the extracted implants. All the detected complications were identified as fractures of the implant. There was no detection of other side effects. From the detected implant fractures, 50% were identified as fissure lines at the implant neck but allowed the successful removal of the implant. The other implant fractures were located at the apical third of the implant body.
Table 1 shows the relevant available information regarding the fractured dental implants. Ten dental implant fractures were observed in 7 patients. There were no differences regarding the location of the fractured implants, as 40% were located in the maxilla and 60% were located in posterior areas (Table 1). Postoperative recovery of the patients was uneventful and pain was successfully managed by oral analgesics. Figures 3 and 4 show one clinical case that was treated with the counter-torque technique to remove failed dental implants.
To our knowledge, this is the study with the largest sample size reporting the performance, in the clinical practice, of the application of counter-torque technique for the removal of nonmobile dental implants. In this retrospective analysis, the counter-torque method demonstrated to be highly predictable, showing a high success rate. The fracture of the implant (0.65% at the implant apical third) was the only detected complication. The principle mechanism of implant explantation was the application of shear stress to break the osseointegration of the implant. Atraumatic implant explantation could be achieved. Besides, this technique demonstrated to be efficient in the removal of nonmobile implants surrounded by different bone levels, ranging from 0–25% to 76–100%.
Due to the low percentage of explantation failures with the counter-torque technique detected in this cohort and the retrospective design of this study, it has not been possible to detect risk factors that could be associated with implant fracture during the explantation or the need to use trephine burs. The influence of the bone quality in the implant removal could not also be assessed. Thus, future prospective studies would be useful to establish a scientifically sound protocol for dental implant explantation using the counter-torque strategy.
The good performance of the counter-torque method, as an atraumatic alternative for nonmobile implant removal, reinforces its use in the clinical practice. However, the technique is not exempt from complications (although at a very low rate).
Availability of data and materials
Data will not be shared but are available upon request.
Howe M-S, Keys W, Richards D. Long-term (10-year) dental implant survival: a systematic review and sensitivity meta-analysis. J Dent. 2019. https://doi.org/10.1016/j.jdent.2019.03.008.
Buser D, Sennerby L, De Bruyn H. Modern implant dentistry based on osseointegration: 50 years of progress, current trends and open questions. Periodontol 2000. 2017;73(1):7–21. https://doi.org/10.1111/prd.12185.
Wang WC, Lagoudis M, Yeh C-W, Paranhos KS. Management of peri-implantitis - a contemporary synopsis. Singap Dent J. 2017;38:8–16. https://doi.org/10.1016/j.sdj.2017.10.001.
Anitua E, Murias-Freijo A, Alkhraisat MH. Conservative implant removal for the analysis of the cause, removal torque, and surface treatment of failed nonmobile dental implants. J Oral Implantol. 2016;42(1):69–77. https://doi.org/10.1563/aaid-joi-D-14-00207.
Solderer A, Al-Jazrawi A, Sahmann P, Jung R, Attin T, Schmidlin P. Removal of failed dental implants revisited: questions and answers. Clin Exo Dent Res. 2019. https://doi.org/10.1002/cre2.234.
Gerlach RC, Dixon DR, Goksel T, Castle JT, Henry WA. Case presentation of florid cemento-osseous dysplasia with concomitant cemento-ossifying fibroma discovered during implant explantation. Oral Surg Oral Med Oral Pathol Oral Radiol. 2013;115(3):e44–52. https://doi.org/10.1016/j.oooo.2012.05.020.
Stajčić Z, Stojčev Stajčić LJ, Kalanović M, Đinić A, Divekar N, Rodić M. Removal of dental implants: review of five different techniques. Int J Oral Maxillofac Surg. 2016;45(5):641–8. https://doi.org/10.1016/j.ijom.2015.11.003.
Anitua E, Orive G. A new approach for atraumatic implant explantation and immediate implant installation. Oral Surg Oral Med Oral Pathol Oral Radiol. 2012;113(3):e19–25. https://doi.org/10.1016/j.tripleo.2011.06.035.
Anitua E, Piñas L, Begoña L, Alkhraisat MH. Prognosis of dental implants immediately placed in sockets affected by peri-implantitis: a retrospective pilot study. Int J Periodontics Restorative Dent. 2012;37(5):713–9. https://doi.org/10.11607/prd.2458.
No funding was received for this study.
Ethics approval and consent to participate
This study was performed following the Helsinki Declaration regarding the investigation with human subjects.
Consent for publication
Eduardo Anitua is the Scientific Director of BTI Biotechnology Institute (Vitoria, Spain). He is the head of the Foundation Eduardo Anitua, Vitoria, Spain. Sofia Fernandez-de-Retana and Mohammad H. Alkhraisat are researchers at BTI Biotechnology Institute (Vitoria, Spain).
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Anitua, E., Fernandez-de-Retana, S. & Alkhraisat, M.H. Performance of the counter-torque technique in the explantation of nonmobile dental implants. Int J Implant Dent 6, 1 (2020) doi:10.1186/s40729-019-0197-z
- Dental implant
- Implant removal
- Minimally invasive